Novel Method for Real Time Co-Registration of IVUS and Coronary Angiography
|The safety and scientific validity of this study is the responsibility of the study sponsor and investigators. Listing a study does not mean it has been evaluated by the U.S. Federal Government. Read our disclaimer for details.|
|ClinicalTrials.gov Identifier: NCT02373826|
Recruitment Status : Unknown
Verified March 2015 by Hillel Yaffe Medical Center.
Recruitment status was: Recruiting
First Posted : February 27, 2015
Last Update Posted : March 10, 2015
|Condition or disease|
|Intravascular Ultrasound (IVUS) Angiography Co-registration|
|Study Type :||Observational|
|Estimated Enrollment :||40 participants|
|Study Start Date :||March 2015|
|Estimated Primary Completion Date :||March 2015|
|Estimated Study Completion Date :||March 2015|
- Accuracy of Co-Registration of IVUS and Coronary Angiography [ Time Frame: One month ]For determination of the accuracy of the co-registration feature, and the accuracy and precision of the length measurements, a flexible phantom was used in order to simulate an artery and enable simulation of the cardiac motion. The sample size for testing the co-registration algorithm was 108 samples (each sample was defined as a landmark). The resultant mean of the absolute distance accuracy was 0.995 mm with a standard deviation of 0.809 mm. The statistical validation in phantoms, resulted in an accuracy of 1.12 mm, meeting the original specification.
To learn more about this study, you or your doctor may contact the study research staff using the contact information provided by the sponsor.
Please refer to this study by its ClinicalTrials.gov identifier (NCT number): NCT02373826
|Contact: Aharon Frimerman, MDfirstname.lastname@example.org|
|Hillel Yaffe Medical Center||Recruiting|
|Hadera, Israel, 38100|
|Contact: Aharon Frimerman, MD 972-52-3543264 email@example.com|